I have a question about the installation of a DP Level Transmitter. Is there any obligation to select both nozzles of a side DP transmitter in the same orientation? I have heard tube length shall be restricted in high temperature services to preventing condensation and we have to select both nozzles in same orientation. Please clarify.
Have an insight or suggestion?
Login or register to post a comment.
Greg McMillan Forum Moderator 36 Posts
Re: Transmitter nozzle orientation27 August 2007 at 1:29pmYou can purge the upper sensing line with a gas compatible with your process (e.g. nitrogen) to keep the line free of condensate. The gas purge flow should be small and constant.